For High Speed Data Acquisition
Transfer of a complete spectral data set consisting of 2048 float values
The BTS2048-VL is controlled through a USB2.0 or Ethernet interface. The data transfer rate using the Ethernet interface is much faster than USB 2.0. The Ethernet interface can be directly connected to a network using a patch cable. The device is allocated an IP address by the network. Direct connection to the PC is also possible; in this case, the device allocates an IP address itself.
Gigahertz-Optik recommends using the USB interface for direct operation of the BTS2048-
VL using a PC since no extra power supply is required. RS232 and RS485 interfaces are also standard. These could be used for an external temperature sensor for example. Two outputs and one input are available for use for triggering functions in automated production applications.
Rear Panel Connector Layout
For complete technical specifications: please visit the Gigahertz-Optik main website here.
Next technical note in series: Maximum Computing Power Using Three Microprocessors